-----BEGIN PGP SIGNED MESSAGE-----
AUSCERT External Security Bulletin Redistribution
Advisory (ICSMA-18-235-01) BD Alaris Plus
24 August 2018
AusCERT Security Bulletin Summary
Product: BD Alaris
Operating System: Network Appliance
Impact/Access: Unauthorised Access -- Remote/Unauthenticated
CVE Names: CVE-2018-14786
- --------------------------BEGIN INCLUDED TEXT--------------------
BD Alaris Plus
All information products included in http://ics-cert.us-cert.gov are
provided "as is" for informational purposes only. The Department of Homeland
Security (DHS) does not provide any warranties of any kind regarding any
information contained within. DHS does not endorse any commercial product or
service, referenced in this product or otherwise. Further dissemination of
this product is governed by the Traffic Light Protocol (TLP) marking in the
header. For more information about TLP, see http://www.us-cert.gov/tlp/.
1. EXECUTIVE SUMMARY
o CVSS v3 9.4
o ATTENTION: Exploitable remotely/low skill level to exploit
o Vendor: Becton, Dickinson and Company (BD)
o Equipment: Alaris GS, Alaris GH, Alaris CC, Alaris TIVA
o Vulnerability: Improper Authentication
2. RISK EVALUATION
Successful exploitation of this vulnerability may allow a remote attacker to
gain unauthorized access to various Alaris Syringe pumps and impact the
intended operation of the pump when it is connected to a terminal server via
the serial port. BD has determined the affected products are not sold within
the United States.
3. TECHNICAL DETAILS
3.1 AFFECTED PRODUCTS
The following versions of Alaris Plus, medical syringe pumps, Versions 2.3.6
and prior, are affected:
o Alaris GS,
o Alaris GH,
o Alaris CC, and
o Alaris TIVA.
3.2 VULNERABILITY OVERVIEW
3.2.1 IMPROPER AUTHENTICATION CWE-287
The software does not perform authentication for functionality that requires a
provable user identity.
CVE-2018-14786 has been assigned to this vulnerability. A CVSS v3 base score
of 9.4 has been calculated; the CVSS vector string is (AV:N/AC:L/PR:N/UI:N/S:U
o CRITICAL INFRASTRUCTURE SECTORS: Healthcare and Public Health
o COUNTRIES/AREAS DEPLOYED: European Union
o COMPANY HEADQUARTERS LOCATION: United States
Elad Luz of CyberMDX discovered this vulnerability, and BD reported this
vulnerability to NCICC.
BD reports that this vulnerability cannot be performed if the device is
connected to an Alaris Gateway Workstation docking station. Also, an attacker
cannot switch the device on remotely, and no PHI or PII can be accessed by
exploiting this vulnerability.
BD recommends the following mitigations and compensating controls in order to
reduce risk associated with this vulnerability:
o This attack utilizes a known vulnerability in terminal servers. Users who
utilize terminal servers should understand that terminal server use is not
o Users should ensure they are operating these devices in a segmented
network environment or as a stand-alone device.
o Users should utilize connections via the Alaris Gateway Workstation
docking station, which would inactivate the remote control feature.
For more information on BD's product security and vulnerability management,
contact their product security office at:
NCCIC recommends users take defensive measures to minimize the risk of
exploitation of this vulnerability. Specifically, users should:
o Minimize network exposure for all control system devices and/or systems,
and ensure that they are not accessible from the Internet.
o Locate control system networks and remote devices behind firewalls, and
isolate them from the business network.
o When remote access is required, use secure methods, such as Virtual
Private Networks (VPNs), recognizing that VPNs may have vulnerabilities
and should be updated to the most current version available. Also
recognize that VPN is only as secure as the connected devices.
NCCIC reminds organizations to perform proper impact analysis and risk
assessment prior to deploying defensive measures.
NCCIC also provides a section for control systems security recommended
practices on the ICS-CERT web page. Several recommended practices are
available for reading and download, including Improving Industrial Control
Systems Cybersecurity with Defense-in-Depth Strategies.
Additional mitigation guidance and recommended practices are publicly
available on the ICS-CERT website in the Technical Information Paper,
ICS-TIP-12-146-01B--Targeted Cyber Intrusion Detection and Mitigation
Organizations observing any suspected malicious activity should follow their
established internal procedures and report their findings to NCCIC for
tracking and correlation against other incidents.
No known public exploits specifically target this vulnerability.
- --------------------------END INCLUDED TEXT--------------------
You have received this e-mail bulletin as a result of your organisation's
registration with AusCERT. The mailing list you are subscribed to is
maintained within your organisation, so if you do not wish to continue
receiving these bulletins you should contact your local IT manager. If
you do not know who that is, please send an email to firstname.lastname@example.org
and we will forward your request to the appropriate person.
NOTE: Third Party Rights
This security bulletin is provided as a service to AusCERT's members. As
AusCERT did not write the document quoted above, AusCERT has had no control
over its content. The decision to follow or act on information or advice
contained in this security bulletin is the responsibility of each user or
organisation, and should be considered in accordance with your organisation's
site policies and procedures. AusCERT takes no responsibility for consequences
which may arise from following or acting on information or advice contained in
this security bulletin.
NOTE: This is only the original release of the security bulletin. It may
not be updated when updates to the original are made. If downloading at
a later date, it is recommended that the bulletin is retrieved directly
from the author's website to ensure that the information is still current.
Contact information for the authors of the original document is included
in the Security Bulletin above. If you have any questions or need further
information, please contact them directly.
Previous advisories and external security bulletins can be retrieved from:
Australian Computer Emergency Response Team
The University of Queensland
Internet Email: email@example.com
Facsimile: (07) 3365 7031
Telephone: (07) 3365 4417 (International: +61 7 3365 4417)
AusCERT personnel answer during Queensland business hours
which are GMT+10:00 (AEST).
On call after hours for member emergencies only.
-----BEGIN PGP SIGNATURE-----
-----END PGP SIGNATURE-----